Terason t3200 Image Transfer Troubleshooting
Verify Network Connection
Find the Wireless icon in the bottom right section of the desktop.
Single Click the icon to see the list of available wireless networks. This image example shows a connection to the MuscleSound wireless network.
Select the desired Network to make changes. (Connect, Disconnect)
Two-finger click (right-click) to bring up more options (Disconnect/Connect, Status, Properties)
Select Properties if you would like to verify the wireless password. Select Status to view signal strength, and other details.
Wired Ethernet Connection
Find the Internet connection icon in the bottom right section of the desktop.
Select 'Network and Sharing Center'
From here you can verify connectivity, change network settings, and have windows troubleshoot problems.
With a successful connection via wireless or ethernet cable, verify internet browser can access the internet
DICOM Server Troubleshooting
Select 'SETUP' on the ultrasound screen.
Select the DICOM tab
Be sure 'DCM4CHEE' is selected in Server dropdown list and Select the 'Edit' button.
Verify these settings... most notably the IP Address: pacs.musclesound.com
If settings are changed, you must OK out of window and select "Apply" to save new settings.
Select 'Verify' to check DICOM server connection. If it shows 'connected', images should transfer to the MuscleSound pacs server.
"TimeOut" the IP Address is probably incorrect.
"Verify Failed", computer doesn't have internet connection.
"Network Service is not running"
Close Ultrasound Program, and Reopen Terason program